Hobart started recording 219/03:04:20. on disks ATNF_V002B_P1
Late getting started due to problem with power supply into VSIC which was oscillating. Took a while to track down as the symptom was no 1PPS or no clock and the VSIC appeared to be OK (as the lights were on).
New, strong RFI at 2147.5 MHz (5 MHz wide) which is out of band but 20-30 dB above the noise at zenith. Looks to be from a tower on a hill to the north. ACA notified. Even though it's out of band, Tsys measurements are affected.