This shows you the differences between two versions of the page.
lbaops:lbaaug2008:v191bholog [2015/12/18 16:38] (current)
|Line 1:||Line 1:|
|+||Hobart started recording 219/03:04:20.|
|+||on disks ATNF_V002B_P1|
|+||Late getting started due to problem with power supply into VSIC which was oscillating. Took a while to track down as the symptom was no 1PPS or no clock and the VSIC appeared to be OK (as the lights were on).|
|+||New, strong RFI at 2147.5 MHz (5 MHz wide) which is out of band but 20-30 dB above the noise at zenith. Looks to be from a tower on a hill to the north. ACA notified. Even though it's out of band, Tsys measurements are affected.|