Recorded data to CURT V006B
Using Agilent @ 13.04 for effective first LO if 8940 MHz.
DAS profile VSOP.PRO
2nd LO 637 for sky freq 8425, 669 for 8457
Coherence in all bandpasses
Pol confirmed with fringe tests on 1921-293.
Bigbuf errors on cDisko between 022157 and 022228
Missing PPS (5/100) lead to restart of recorder @ 0551
Windstowed @0548-0555 0724-0732 0737-0744
Ceduna maser suffered a glitch at ~0100/58, causing a change in the tick phase derivative.