|Pass||File name||Description||Start date, UT range||Antennas||Polarizations||# subbands (AIPS IFs)||Bandwidth per IF (MHz)||Spectral channels per IF/pol||Corr. int. time (s)|
|v596b||V596B.FITS||2021-07-18, 0/16:55:16 - 1/11:55:15||AT CD HO KE MP PA TI WW YG||RR LL RL LR||4||16.0 MHz||32||1.0|
AT: Small clock jumps (~20 nsec) at ~23:12 and ~04:39. Phase/delay solutions will not transfer across these boundaries. The tied array was also accidentally formed with atmospheric corrections applied - this has been compensated in the correlator but might result in slightly larger phase referencing residuals than normal.
TI: DSS34, RCP only. 07:22 to 10:22
KE: Subband 1-2 v. low sensitivity. Subband 3-4 RCP Low sensitivity. Data could not be pol converted (still in linear basis).
PA: Polarization basis converted from linear to circular *after* correlation.
This is still experimental, so please inspect data carefully.
Plots of autocorrelations
Plots of cross-polarization amplitude and phase against frequency (not always available)
Amplitude corrections from ACCOR
Fringe-fit delay solutions
Fringe-fit phase solutions
Fringe-fit rate solutions